Mildly alkaline liquid cleaner for use in cleaning and disinfection devices Residue-free, thorough cleaning with very good material compatibility as well as against color anodized aluminum. Free from surfactants and thus foam-free.
|Our performance||Your advantage|
|Specially developed formulation with high material protection||
Very good material compatibility even against sensitive materials e.g. (color-) anodized aluminum
|Surfactant-free with high performance formulation components||Low foam development and very good cleaning performance against typical stains from the medical area|
Addition of corrosion inhibitors
|Broad use spectrum over almost all materials used in the medical area|
|pH > 10 in the application solution at 55 ºC||Secure reprocessing of instruments at all water hardness grades|
In cleaning and disinfection devices in the medical area, depending on the water quality, can be used with 2-8ml/L preferably with DI water or softened water.
For surgical instruments and other medical instruments and accessories that can be thermally disinfected at 93 °C.
For reprocessing OP shoes and thermolabile medical materials at 45 to 60 °C.
Also suitable for reprocessing lab glass and baby bottles.
Chemical thermal disinfection; with Sekumatic™ FD or Sekumatic™ FDR at 60 °C for 5-10 minutes.
When neutralization is needed (bad water quality) Sekumatic™ FNZ or Sekumatic™ FNP can be used.
Thermal disinfection: For final rinsing, preferably use fully demineralized water (DI).
Sekumatic™ MultiClean has phosphates, silicates, potassium hydroxide and corrosion inhibitors as the base.
|In container1||Until the printed expiration date|
Please note the following to ensure the correct storage of environmental hygiene and instrument disinfectants: The original containers (canisters, buckets or bottles) with the disinfectant concentrate (or powder) should be stored at temperatures below 25°C and kept dry and clean. A brief period below or above the given storage temperature (e.g. during transport) will not diminish the effectiveness.
The containers are to be sealed tightly after each use. Please also take care that the concentrate remaining in the container is not contaminated when preparing the application solution.
|Chemical & Physical Data||Temperature||Value|
|Density||20 °C||1.3 g / ml|
|pH value* (concentrate)||20 °C||ca. 13.4|
|pH-range* (solution, 5-8ml/l in DI-water)||20 °C||11.3 – 11.6|
|Conductivity (solution, 5-8ml/l in DI-water)||20 °C||2.2 – 3.2 mS / cm|
* pH-value measurements under operating conditions always by using pH-meter with temperature compensation
Instrument cleaning of medical devices
Please see below related offerings:
Our search returned no results.